The Mikro Air Jet Sieve – MAJSx has the capability to print
test results straight from the Mikro Air Jet Sieve. This will ensure that the test results from
tests on your MAJSx are produced and distributed speedily and with ease.
The instructions for setting up your printer with your MAJSx
are as follows:
Ensure you have a printer
that is compatible with a PDF language. A list of compatible printers has
Connect the Mikro Air Jet
Sieve to a network. You may do this through your corporate network by
plugging an Ethernet cable into the Ethernet port on the back of the Mikro
Air Jet Sieve. You may also use a mini broadband router, connecting a
router to both the Mikro Air Jet Sieve and the printer.
Identify the IP address on
Power up your air jet
sieve and go to the “options” screen. Look for the IP address in the
“Printer IP” bar. It should correspond with the IP address of your
printer. If it is the correct IP address, hit “save.” A window should pop
up saying that the IP address was saved. Click on “OK.”
After conducting a test,
hit the print button, and it should print to your printer. Note: You may
want to back up your data by hitting “save” on the same screen that you
hit “Print.” Also, in order to print, you MUST have a USB stick inserted
into one of the USB ports.
After ensuring your data
is saved, you may hit “Print,” and it should send a signal to your printer
via IP address, telling your printer to print your test results.
Once the analysis is
printed, you may access a copy of the printout on the USB stick.
Note: There are two types of print modes, Lin Lin, and Rosin
Rammler. Lin Lin allows the user to print a graph with a test that used only
one test screen. Rosin Rammler is a log-lin graph that can only prints tests
that used 2 or more test screens to print.
Note: You can only use spaces in the “notes” section. In all
other sections, you may substitute underscores for spaces.